Training and Support for the Breath Hold ES
Click on tabs to download training materials for the Breath Hold ES
Come visit us at the North American Society for Pediatric Gastroenterology, Hepatology and Nutrition (NASPGHAN) Annual Meeting, Booth# 101 Diplomat Hotel, Hollywood FL October 25th – 27th, 2018 About NASPGHAN NASPGHAN is the only professional society for pediatric...
Come visit us at the Texas Society of Gastroenterology and Endoscopy (TSGE) Annual Meeting, Booth #48 Austin Hilton, Austin TX September 21st – 23rd, 2018 About TSGE Texas Society for Gastroenterology and Endoscopy is a statewide professional association for...
Come visit us the Florida Gastroenterologic Society (FGS) Annual Meeting, Booth #324 Hyatt Regency Grand Cypress, Orlando FL September 14th – 16th, 2018 About FGS FGS strives to be the organization of choice for Florida Gastroenterologist. We work for members in...
Come visit us at the American Urogynecologic Society (AUGS) PFD Week, Booth #308 Hyatt Regency Chicago, Chicago IL October 9th – 13th About AUGS PFD Week 2018 PFD Week was created by AUGS to become the go-to meeting for health care professionals interested in or...
What systems are the Breath Hold ES compatible with?
The Breath Hold ES is a portable, standalone system that acts independently from any equipment. Allowing it to be used with any linac, MRI, CT etc.
What does the Breath Hold ES accomplish?
It restricts target tissue mobility through consistent breath-holds or controlled shallow breathing.
How does the Breath Hold ES work?
It relies on biofeedback for proactive patient breathing control.
Where does the Bellows Belt go on the patient?
It goes around the chest or abdomen for greater patient comfort and ease of use than other radiation therapy respiratory motion control technologies.
What sort of maintenance for the Bellows Belt is required?
You just need to clean the belt with propyl alcohol and that will allow you to reuse the built between patients.
How does the belt communicate with the Biofeedback?
There are pressure sensitive transducers that read the circumference of the belt and transmit the findings in real time to the display screens.
Are there any materials that could conflict with any machines?
The Breath Hold ES is made from entirely MRI safe materials and the radio signal will not disrupt any Linac, MRI, etc.
Quick Reference Guide: