BC-10 Coil FAQ
These are the most common FAQ regarding the BC-10 coil.
Some images in a 3 plane scout study are nothing but noise?
It is possible that one plane of the scout has missed the object. If one or both of the other planes contain good images, the coil is operating normally.
I have installed the Medspira BC-10 Transmit/Receive MRI Coil onto my scanner – how do I know it is working properly?
1. Run the “Coil Imaging Performance Verification Check” procedure outlined in section 3-2 of your Owner’s Manual or Service Manual. If the coil does not meet specifications continue to step B.
2. The magnet polarity on your MRI system may not be correct for the default cable configuration of the BC-10 MRI Coil. To check this on a GE Healthcare MRI system perform the following – use manual prescan and set the TG to 20 (this is a very low power value) while viewing the coarse center frequency display. If a central peak is shown that extends well above the baseline – the coil is wired properly. If not, the connector must be opened by technical personnel and the wires swapped. Repeat step A to confirm proper coil installation. If the BC-10 MRI coil still does not meet specifications contact Medspira for technical assistance.
My images appear quite noisy?
Double check the imaging protocol and compare the images to previous ones you may have acquired. If a coil problem is still suspected, conduct the SNR test listed in the operator’s manual and compare the measured coil SNR to the specifications.
Can I leave the coil in the bore while imaging with the body coil?
No – imaging using the body coil with the Transmit/Receive MRI Coil in the bore will damage the Medspira Transmit/Receive MRI Coil.
What does a T/R coil do that is so special?
T/R or transmit – receive coils transmit locally using (in our case) the Medspira Transmit/Receive MRI Coil (not the body coil). This uses very low RF power and localizes the RF field essentially to the inside of the Medspira Transmit/Receive MRI Coil. This will minimize signals from outside the coil and so can be used at the patient’s side without forcing phase encoding or frequency encoding onto specific directions. Also the local RF is very uniform with this design – so fat saturation is often quite uniform (however keep in mind that as the patients’ part of interest is moved away from isocenter – the main magnetic field is not very uniform and fat saturation techniques are difficult).
American Society of Colon & Rectal Surgeons 2019 Come visit us at the American Society of Colon & Rectal Surgeons (ASCRS) Annual Scientific Meeting, Booth #727 Huntington Convention Center of Cleveland, Cleveland OH June 1st – 5th, 2019 About ASCRS With emphasis on...
Digestive Disease Week 2019 Come visit us at Digestive Disease Week (DDW), Booth #3321 San Diego Convention Center, San Diego CA, May 18th – 21st, 2019 About DDW Recognized as one of the top 50 medical meetings by HCEA, Digestive Disease Week® (DDW) is the world's...
Come visit us at the upcoming New York Society for Gastrointestinal Endoscopy (NYSGE) Annual Course, Booth # 608. New York Marriott Marquis Hotel, New York NY December 13th – 14th, 2018. About NYSGE The New York Society for Gastrointestinal Endoscopy (NYSGE) was...
Come visit us at the North American Society for Pediatric Gastroenterology, Hepatology and Nutrition (NASPGHAN) Annual Meeting, Booth# 101 Diplomat Hotel, Hollywood FL October 25th – 27th, 2018 About NASPGHAN NASPGHAN is the only professional society for pediatric...