Some images in a 3 plane scout study are nothing but noise?
It is possible that one plane of the scout has missed the object. If one or both of the other planes contain good images, the coil is operating normally.
I have installed the Medspira BC-10 Transmit/Receive MRI Coil onto my scanner – how do I know it is working properly?
1. Run the “Coil Imaging Performance Verification Check” procedure outlined in section 3-2 of your Owner’s Manual or Service Manual. If the coil does not meet specifications continue to step B.
2. The magnet polarity on your MRI system may not be correct for the default cable configuration of the BC-10 MRI Coil. To check this on a GE Healthcare MRI system perform the following – use manual prescan and set the TG to 20 (this is a very low power value) while viewing the coarse center frequency display. If a central peak is shown that extends well above the baseline – the coil is wired properly. If not, the connector must be opened by technical personnel and the wires swapped. Repeat step A to confirm proper coil installation. If the BC-10 MRI coil still does not meet specifications contact Medspira for technical assistance.
My images appear quite noisy?
Double check the imaging protocol and compare the images to previous ones you may have acquired. If a coil problem is still suspected, conduct the SNR test listed in the operator’s manual and compare the measured coil SNR to the specifications.
Can I leave the coil in the bore while imaging with the body coil?
No – imaging using the body coil with the Transmit/Receive MRI Coil in the bore will damage the Medspira Transmit/Receive MRI Coil.
What does a T/R coil do that is so special?
T/R or transmit – receive coils transmit locally using (in our case) the Medspira Transmit/Receive MRI Coil (not the body coil). This uses very low RF power and localizes the RF field essentially to the inside of the Medspira Transmit/Receive MRI Coil. This will minimize signals from outside the coil and so can be used at the patient’s side without forcing phase encoding or frequency encoding onto specific directions. Also the local RF is very uniform with this design – so fat saturation is often quite uniform (however keep in mind that as the patients’ part of interest is moved away from isocenter – the main magnetic field is not very uniform and fat saturation techniques are difficult).
Come visit us at ASCRS 2018 at booth #623 May 19- 23, 2018 Music City Center, Nashville, TN Join us for the 2018 Annual Meeting at: Music City Center Omni Nashville Hotel Nashville, Tennessee May 19 – 23, 2018 The Annual Meeting features new, innovative...
Come visit us at NASPGHAN 2017 at Booth #225 Ceaser’s Palace, November 1-4, 2017 NASPGHAN Postgraduate Course
Come visit us at World Congress of Gastroenterology @ ACG2017 at Booth #429 Orange Town Convention Center, October…
Come visit us at PFD Week 2017 at Booth #615 October, 2017 Providence, RI Tuesday, October 3 7:00 am – 5:00 pm Registration Open 8:30 am – 12:00 pm Pre-Conference Workshops Untitled_design_(12) 1:30 pm – 5:00 pm Pre-Conference Workshops...